Please help us to find bad videos. Broken or unappropriated video content?
In this episode Shahriar repairs an Agilent 54845A oscilloscope with an intermittent failure. The instrument occasionally fails self-calibration and the displayed waveform shows undesired spurious activities. The teardown of the instrument reveals customized front-end section coupled into a ceramic-substrate ADC ASIC designed by Agilent. The packaged ADCs are placed in sockets. The ICs are removed and the ceramic carrier is cleaned; this resolves the intermittent problem with the channels. The architecture of the scope is also presented. The functionality of the scope is verified through self-calibration, self-test and measurement of various waveform.
The Signal Path
Impresspages.lt is not the owner of this text/video/image/photo content, the real source of content is Youtube.com and user declared in this page publication as Youtube.com user,
if you have any question about video removal, what was shared by open community, please contact Youtube.com directly or report bad/not working video links directly to video owner on Youtube.com. Removed video from Youtube.com will also be removed from here.